What Is Karl Fischer Titration?
Karl Fischer titration is a titration method which uses volumetric or coulometric titration to determine the trace quantity of water in the sample. It was invented by a German chemist by name Karl Fischer in the year 1935.
Karl Fischer Titration Principle
This experiment is based on iodine/iodide reaction. Water reacts with iodine and reaches endpoint when all the water is consumed. This method uses an organic base (B), alcohol, iodine, and sulfur dioxide. In the original KF experiment, the alcohol used was methanol and organic base used was pyridine. During the experiment and course of the titration, the sample is added with iodine and the amount of iodine that is used to consume all the water in the sample is measured.
Karl Fischer Titration Equipment
Drying tube, sample injection cap, electrode analysis, Drain cook, a cathode chamber, detection electrode, rotor, anode chamber, KF reagent.
Ingredients of KF reagent:
Iodine, Buffer (Imidazole), sulfur dioxide, solvent (methanol).
Karl Fischer Titration Procedure
The Karl Fischer titration experiment can be performed in two different methods. They are:
- Volumetric determination – This technique is suitable to determine water content down to 1% of water. The sample is dissolved in KF methanol and the iodine is added to KF Reagent. The endpoint is detected potentiometrically.
- Coulometric determination – The endpoint is detected in this experiment electrochemically. Iodine required for KF reaction is obtained by anodic oxidation of iodide from solution.
Karl Fischer Titration Applications
- It is used in technical products such as plastics, oils, gases.
- It is used in pharmaceutical products.
- It is used in cosmetic products.
- It is used in the industry.
Advantages of Karl Fischer Titration
- It is fitted for determining water in gases, liquids and solids.
- The coulometric titrator helps in detecting free water, dissolved water, and emulsified water.
- It is a swift process which demands a minimal amount of sample preparation.
- Extremely accurate method.
Limitations of Karl Fischer Titration
- It is a destructive technique.
- The solvent consumption is high as the manual volumetric titration demands reloading during each determination.
- Coulometric titration is fitted only for samples that contain a small amount of water.
- Coulometric titration takes extremely long periods to determine.
To learn more about the Karl Fischer Titration from the experts at BYJU’S, register now!
Frequently Asked Questions
What is the use of methanol in Karl Fischer titration?
Methanol is used as a solvent.
How to prepare KF reagent?
Make a solution by mixing 170 mL of pyridine and 670 mL of methanol. Add 125 g of iodine to the solution and cool it. Take a 250 mL graduated cylinder and add 100 mL of pyridine. Keep it on an ice bath. Pass in sulfur dioxide (dry) till its volume reaches 200 mL.
What is a major difference between volumetric and coulometric titration?
The main difference between them is that:
Volumetric method – the titrant is directly added to the sample with the help of a burette.
Coulometric method – the titrant is produced electrochemically inside the titration cell.
What is coulometric Karl Fischer?
The Karl Fischer titration is merely a means of measuring sample water content. Modern instruments, such as the Aquamax KF, apply the coulometric principle, whereby the water present in the sample is coulometrically titrated to a predefined endpoint at which free iodine exceeds a minute.
Why we use sodium tartrate in Karl Fischer?
The volumetric standard for Karl Fischer titration is sodium tartrate dihydrate. It is stable and non-hygroscopic, under normal conditions. Sodium tartrate dihydrate has a 15.66 percent stoichiometric water content and is primarily used in volumetry to measure the titer.
Other important links: